IonPumps/current/dbandHi = 1.6
IonPumps/current/dbandLo = 0.0
IonPumps/current/limitHi = 3.2
IonPumps/current/limitLo = 0.0
IonPumps/voltage/dbandHi = 500.0
IonPumps/voltage/dbandLo = 0.0
IonPumps/voltage/limitHi = 7000.0
IonPumps/voltage/limitLo = 0.0
VQMonitor/vqmpressure/dbandHi = 20.0
VQMonitor/vqmpressure/dbandLo = 1.0E-8
VQMonitor/vqmpressure/limitHi = 810.0
VQMonitor/vqmpressure/limitLo = 0.0
cold1/temp_Min_OFF = -115.0
cold1/temp_Min_ON = -75.0
cold1/autoOffEnabled/dbandHi = 0.0
cold1/autoOffEnabled/dbandLo = 0.0
cold1/autoOffEnabled/limitHi = 0.5
cold1/autoOffEnabled/limitLo = 0.0
cold1/autoOffTemperature/dbandHi = 0.0
cold1/autoOffTemperature/dbandLo = 0.0
cold1/autoOffTemperature/limitHi = 25.0
cold1/autoOffTemperature/limitLo = -150.0
cold1/autoOnTemperature/dbandHi = 0.0
cold1/autoOnTemperature/dbandLo = 0.0
cold1/autoOnTemperature/limitHi = 25.0
cold1/autoOnTemperature/limitLo = -150.0
cold1/power/dbandHi = 0.0
cold1/power/dbandLo = 0.0
cold1/power/limitHi = 250.0
cold1/power/limitLo = 0.0
cold1/rejectTemperature/dbandHi = 0.0
cold1/rejectTemperature/dbandLo = 0.0
cold1/rejectTemperature/limitHi = 25.0
cold1/rejectTemperature/limitLo = -150.0
cold1/setPower/dbandHi = 0.0
cold1/setPower/dbandLo = 0.0
cold1/setPower/limitHi = 250.0
cold1/setPower/limitLo = 0.0
cold1/setpoint/dbandHi = 0.0
cold1/setpoint/dbandLo = 0.0
cold1/setpoint/limitHi = 28.0
cold1/setpoint/limitLo = -130.0
cold1/temperature/dbandHi = 0.0
cold1/temperature/dbandLo = 5.0
cold1/temperature/limitHi = 28.0
cold1/temperature/limitLo = -115.0
cold2/temp_Min_OFF = -115.0
cold2/temp_Min_ON = -75.0
cold2/autoOffEnabled/dbandHi = 0.0
cold2/autoOffEnabled/dbandLo = 0.0
cold2/autoOffEnabled/limitHi = 0.5
cold2/autoOffEnabled/limitLo = 0.0
cold2/autoOffTemperature/dbandHi = 0.0
cold2/autoOffTemperature/dbandLo = 0.0
cold2/autoOffTemperature/limitHi = 25.0
cold2/autoOffTemperature/limitLo = -150.0
cold2/autoOnTemperature/dbandHi = 0.0
cold2/autoOnTemperature/dbandLo = 0.0
cold2/autoOnTemperature/limitHi = 25.0
cold2/autoOnTemperature/limitLo = -150.0
cold2/power/dbandHi = 0.0
cold2/power/dbandLo = 0.0
cold2/power/limitHi = 250.0
cold2/power/limitLo = 0.0
cold2/rejectTemperature/dbandHi = 0.0
cold2/rejectTemperature/dbandLo = 0.0
cold2/rejectTemperature/limitHi = 25.0
cold2/rejectTemperature/limitLo = -150.0
cold2/setPower/dbandHi = 0.0
cold2/setPower/dbandLo = 0.0
cold2/setPower/limitHi = 250.0
cold2/setPower/limitLo = 0.0
cold2/setpoint/dbandHi = 0.0
cold2/setpoint/dbandLo = 0.0
cold2/setpoint/limitHi = 28.0
cold2/setpoint/limitLo = -130.0
cold2/temperature/dbandHi = 0.0
cold2/temperature/dbandLo = 0.0
cold2/temperature/limitHi = 28.0
cold2/temperature/limitLo = -95.0
cryo/temp_Min_OFF = -170.0
cryo/temp_Min_ON = -120.0
cryo/autoOffEnabled/dbandHi = 0.0
cryo/autoOffEnabled/dbandLo = 0.0
cryo/autoOffEnabled/limitHi = 2.0
cryo/autoOffEnabled/limitLo = 0.0
cryo/autoOffTemperature/dbandHi = 0.0
cryo/autoOffTemperature/dbandLo = 0.0
cryo/autoOffTemperature/limitHi = 25.0
cryo/autoOffTemperature/limitLo = -170.0
cryo/autoOnTemperature/dbandHi = 0.0
cryo/autoOnTemperature/dbandLo = 0.0
cryo/autoOnTemperature/limitHi = 25.0
cryo/autoOnTemperature/limitLo = -150.0
cryo/power/dbandHi = 0.0
cryo/power/dbandLo = 0.0
cryo/power/limitHi = 250.0
cryo/power/limitLo = 0.0
cryo/rejectTemperature/dbandHi = 0.0
cryo/rejectTemperature/dbandLo = 0.0
cryo/rejectTemperature/limitHi = 25.0
cryo/rejectTemperature/limitLo = -150.0
cryo/setPower/dbandHi = 0.0
cryo/setPower/dbandLo = 0.0
cryo/setPower/limitHi = 250.0
cryo/setPower/limitLo = 0.0
cryo/setpoint/dbandHi = 0.0
cryo/setpoint/dbandLo = 0.0
cryo/setpoint/limitHi = 28.0
cryo/setpoint/limitLo = -170.0
cryo/temperature/dbandHi = 0.0
cryo/temperature/dbandLo = 0.0
cryo/temperature/limitHi = 28.0
cryo/temperature/limitLo = -160.0
rtds/temperatureCold1/dbandHi = 0.0
rtds/temperatureCold1/dbandLo = 1.0
rtds/temperatureCold1/limitHi = 25.0
rtds/temperatureCold1/limitLo = -65.0
rtds/temperatureCold2/dbandHi = 0.0
rtds/temperatureCold2/dbandLo = 1.0
rtds/temperatureCold2/limitHi = 25.0
rtds/temperatureCold2/limitLo = -65.0
rtds/temperatureCryo/dbandHi = 0.0
rtds/temperatureCryo/dbandLo = 2.0
rtds/temperatureCryo/limitHi = 25.0
rtds/temperatureCryo/limitLo = -135.0
turbo/cntrlrAirTemperature/dbandHi = 5.0
turbo/cntrlrAirTemperature/dbandLo = 5.0
turbo/cntrlrAirTemperature/limitHi = 35.0
turbo/cntrlrAirTemperature/limitLo = 0.0
turbo/cntrlrSinkTemperature/dbandHi = 5.0
turbo/cntrlrSinkTemperature/dbandLo = 5.0
turbo/cntrlrSinkTemperature/limitHi = 35.0
turbo/cntrlrSinkTemperature/limitLo = 0.0
turbo/current/dbandHi = 0.0
turbo/current/dbandLo = 0.0
turbo/current/limitHi = 0.0
turbo/current/limitLo = 0.0
turbo/driveFrequency/dbandHi = 0.0
turbo/driveFrequency/dbandLo = 0.0
turbo/driveFrequency/limitHi = 0.0
turbo/driveFrequency/limitLo = 0.0
turbo/power/dbandHi = 31.0
turbo/power/dbandLo = 0.0
turbo/power/limitHi = 71.0
turbo/power/limitLo = 0.0
turbo/pumpTemperature/dbandHi = 5.0
turbo/pumpTemperature/dbandLo = 5.0
turbo/pumpTemperature/limitHi = 35.0
turbo/pumpTemperature/limitLo = 0.0
turbo/rpm/dbandHi = 0.0
turbo/rpm/dbandLo = 0.0
turbo/rpm/limitHi = 0.0
turbo/rpm/limitLo = 0.0
turbo/status/dbandHi = 0.0
turbo/status/dbandLo = 5.0
turbo/status/limitHi = 5.5
turbo/status/limitLo = -0.5
turbo/voltage/dbandHi = 0.0
turbo/voltage/dbandLo = 0.0
turbo/voltage/limitHi = 0.0
turbo/voltage/limitLo = 0.0
