import org.lsst.ccs.description.groovy.CCSBuilder
import org.lsst.ccs.subsystem.power.ATSPower
import org.lsst.ccs.subsystem.power.HamegDevice;
import org.lsst.ccs.subsystem.power.N6487Device;
import org.lsst.ccs.subsystem.power.PowerControl
import org.lsst.ccs.monitor.Channel

CCSBuilder builder = ["ats-power"]

builder.main (ATSPower) {

    def hamegB = "HamegBottom"
    def hamegM = "HamegMiddle"
    def hamegT = "HamegTop"
    def n6487 = "Keithley"

    "$hamegB" (HamegDevice, connType:"serial",
               devcId:"/dev/serial/by-id/usb-HAMEG_HAMEG_HO720_100030280932-if00-port0") {

        "Digital" (PowerControl, description:"Digital voltage", hwChan:2, voltage:5.0, current:1.0, onDelay:0.0, offDelay:0.6)
    }

    "$hamegM" (HamegDevice, connType:"serial",
               devcId:"/dev/serial/by-id/usb-HAMEG_HAMEG_HO720_100030867193-if00-port0") {

        "ClockLow" (PowerControl, description:"Clock low voltage", hwChan:1, voltage:11.0, current:0.5, onDelay:0.3, offDelay:0.3)

        "ClockHigh" (PowerControl, description:"Clock high voltage", hwChan:2, voltage:10.0, current:0.5, onDelay:0.2, offDelay:0.4)

        "Analog"  (PowerControl, description:"Analog voltage", hwChan:3, voltage:7.0, current:1.0, onDelay:0.1, offDelay:0.5)
    }

    "$hamegT" (HamegDevice, connType:"serial",
               devcId:"/dev/serial/by-id/usb-HAMEG_HAMEG_HO720_100030785832-if00-port0") {

        "DPHI" (PowerControl, description:"DPHI voltage", hwChan:1, voltage:10.0, current:1.0, onDelay:0.5, offDelay:0.1)

        "OD" (PowerControl, description:"OD voltage", hwChan:2, voltage:28.0, current:0.5, onDelay:0.4, offDelay:0.2)

        "Heater" (PowerControl, description:"Heater voltage", hwChan:3, voltage:12.0, current:0.5, onDelay:0.6, offDelay:0.0)
    }

    "$n6487" (N6487Device, connType:"serial", devcParm:9600,
              devcId:"/dev/serial/by-id/usb-FTDI_FT232R_USB_UART_A10747HQ-if00-port0") {

        "HVBias" (PowerControl, description:"HV bias voltage", hwChan:0, voltage:24.0, current:0.001, onDelay:0.0, offDelay:0.0)
    }

    "Digital_V" (Channel, description: "Digital voltage", units: "Volts",
                 devcName: "$hamegB",  hwChan:2, type: "VOLTAGE")
                          
    "Digital_I" (Channel, description: "Digital current", units: "mA", format: ".1f",
                 devcName: "$hamegB",  hwChan:2, type: "CURRENT", scale: 1000.0)
                          
    "Analog_V" (Channel, description: "Analog voltage", units: "Volts",
                devcName: "$hamegM",  hwChan:3, type: "VOLTAGE")
                          
    "Analog_I" (Channel, description: "Analog current", units: "mA", format: ".1f",
                devcName: "$hamegM",  hwChan:3, type: "CURRENT", scale: 1000.0)

    "ClkHigh_V" (Channel, description: "Clock high voltage", units: "Volts",
                 devcName: "$hamegM",  hwChan:2, type: "VOLTAGE")
                          
    "ClkHigh_I" (Channel, description: "Clock high current", units: "mA", format: ".1f",
                 devcName: "$hamegM",  hwChan:2, type: "CURRENT", scale: 1000.0)
                          
    "ClkLow_V" (Channel, description: "Clock low voltage", units: "Volts",
                devcName: "$hamegM",  hwChan:1, type: "VOLTAGE")
                          
    "ClkLow_I" (Channel, description: "Clock low current", units: "mA", format: ".1f",
                devcName: "$hamegM",  hwChan:1, type: "CURRENT", scale: 1000.0)
                          
    "OD_V" (Channel, description: "OD voltage", units: "Volts",
            devcName: "$hamegT",  hwChan:2, type: "VOLTAGE")
                          
    "OD_I" (Channel, description: "OD current", units: "mA", format: ".1f",
            devcName: "$hamegT",  hwChan:2, type: "CURRENT", scale: 1000.0)
                          
    "DPHI_V" (Channel, description: "DPHI voltage", units: "Volts",
              devcName: "$hamegT",  hwChan:1, type: "VOLTAGE")
                          
    "DPHI_I" (Channel, description: "DPHI current", units: "mA", format: ".1f",
              devcName: "$hamegT",  hwChan:1, type: "CURRENT", scale: 1000.0)
     
    "Heater_V" (Channel, description: "Heater voltage", units: "Volts",
                devcName: "$hamegT",  hwChan:3, type: "VOLTAGE")
                          
    "Heater_I" (Channel, description: "Heater current", units: "mA", format: ".1f",
                devcName: "$hamegT",  hwChan:3, type: "CURRENT", scale: 1000.0)
     
    "HVBias_V" (Channel, description: "HV bias voltage", units: "Volts",
                devcName: "$n6487",  hwChan:0, type: "VOLTAGE")
                          
    "HVBias_I" (Channel, description: "HV bias current", units: "mA", format: ".1f",
                devcName: "$n6487",  hwChan:0, type: "CURRENT", scale: 1000.0)
     
}
